Blood Glucose Readings
Making Sure Your Blood Glucose Readings Are Accurate
You should perform quality-control checks to make sure that your blood glucose readings are accurate and reliable. Several things can reduce the accuracy of your meter reading even if it appears to still work. For instance, the meter may have been dropped or its electrical components may have worn out. Humidity or heat may damage test strips. It is even possible that your testing technique may have changed slightly. Quality control checks should be done on a regular basis according to the meter manufacturer's instructions. There are two kinds of quality control checks:
Check Using "Test Quality Control Solutions" or "Electronic Controls". Test quality control solutions and electronic controls are both used to check the operation of your meter and your blood glucose readings. Test quality control solutions check the accuracy of the meter and test strip. They may also give an indication of how well you use your system. Electronic controls only check that the meter is working properly.
Test quality control solutions have known glucose values. Essentially, when you run a quality control test, you substitute the test solution for blood. The difference is that you know what the result should be.
To test your meter with a quality control solution, follow the instructions that accompany the solution. These will guide you to place a certain amount of solution on your test strip and run it through your meter. The meter will give you a reading for the amount of glucose in the sample. Compare this number to the number listed on the control solution range on the bottle of strips. If the results of your test match the values given in the quality control solution labeling, you can be assured the entire system (meter and test strip) is working properly. If results are not correct, the system may not be accurate—contact the manufacturer for advice.
Manufacturers sometimes include quality control solution with their meter. However, most often you must order it separately from a manufacturer or pharmacy.
Some glucose meters also use electronic controls to make sure the meter is working properly. With this method, you place a cartridge or a special "control" test strip in the meter and a signal will appear to indicate if the meter is working.
Take Your Meter with You to The Health Care Provider's Office. This way you can check your blood glucose readings while your health care provider watches your technique to make sure you are using the meter correctly. Your healthcare provider will also take a sample of blood and evaluate it using a routine laboratory method. If values obtained on the glucose meter match the laboratory method, you and your healthcare provider will see that your meter is working well and that you are using good technique. If results do not match the laboratory method results, then results you get from your meter may be inaccurate and you should discuss the issue with your healthcare provider and contact the manufacturer if necessary.
Reviewed by Francine Kaufman, MD. 4/08
Carrot Snack Cake Crabmeat Mold Raspberry Marinated Tuna Steaks Cranberry Nut Muffins Zangria-Style Iced Tea Mixed Greens with Sun-Dried Tomato Vinaigrette Apricot Bars Creamy Low-Fat Pasta With Pimientos Seafood Spread Chicken Cacciatore Dish